<table class="table table-bordered">
 <tbody>
 <tr>
 <td rowspan="2">
 Mass Flow Ranges
 </td>
 <td>
 Model 8803-1:&nbsp;&nbsp; 5-1500 g/min*
 </td>
 </tr>
 <tr>
 <td>
 Model 8803-2:&nbsp;&nbsp; 5-3500 g/min*
 </td>
 </tr>
 <tr>
 <td rowspan="2">
 Zero Offset Stability (Z.O.S.)
 </td>
 <td>
 Model 8803-1:&nbsp;&nbsp; 0.06 g/min
 </td>
 </tr>
 <tr>
 <td>
 Model 8803-2:&nbsp;&nbsp; 0.50 g/min
 </td>
 </tr>
 <tr>
 <td>
 Accuracy
 </td>
 <td>
 ± {1% of rate + Z.O.S.}
 </td>
 </tr>
 <tr>
 <td>
 Temperature
 </td>
 <td>
 Ambient: 0°–50°C; Fluid 15°–40°C
 </td>
 </tr>
 <tr>
 <td>
 Operating Pressure
 </td>
 <td>
 80 PSIG (Max.)
 </td>
 </tr>
 </tbody>
 </table>
 
 
  <h4>Measurement Specifications</h4><h4>
 <table class="table table-bordered">
 <tbody>
 <tr>
 <td rowspan="2">
 Mass Flow Ranges
 </td>
 <td>
 Model 8803-1:&nbsp;&nbsp; 5-1500 g/min*
 </td>
 </tr>
 <tr>
 <td>
 Model 8803-2:&nbsp;&nbsp; 5-3500 g/min*
 </td>
 </tr>
 <tr>
 <td rowspan="2">
 Zero Offset Stability (Z.O.S.)
 </td>
 <td>
 Model 8803-1:&nbsp;&nbsp; 0.06 g/min
 </td>
 </tr>
 <tr>
 <td>
 Model 8803-2:&nbsp;&nbsp; 0.50 g/min
 </td>
 </tr>
 <tr>
 <td>
 Accuracy
 </td>
 <td>
 ± {1% of rate + Z.O.S.}
 </td>
 </tr>
 <tr>
 <td>
 Temperature
 </td>
 <td>
 Ambient: 0°–50°C; Fluid 15°–40°C
 </td>
 </tr>
 <tr>
 <td>
 Operating Pressure
 </td>
 <td>
 80 PSIG (Max.)
 </td>
 </tr>
 </tbody>
 </table>
 *Pressure drop at Max. Flow range = 10 psi on water (1cP)
 &nbsp;
 &nbsp;
 </h4><h4>Electrical Specifications</h4><h4>
 <table class="table table-bordered">
 <tbody>
 <tr>
 <td>
 Supply Voltage
 </td>
 <td>
 24VDC ±10%
 </td>
 </tr>
 <tr>
 <td>
 Power Consumption
 </td>
 <td>
 Max 6W
 </td>
 </tr>
 <tr>
 <td>
 Programming
 </td>
 <td>
 Operator Parameter configuration through USB interface with a PC
 </td>
 </tr>
 <tr>
 <td>
 Output Interfaces
 </td>
 <td>
 4-20 mA Current Loop, Digital I/O
 </td>
 </tr>
 <tr>
 <td>
 LCD Display
 </td>
 <td>
 2 lines; 16 characters per line
 </td>
 </tr>
 <tr>
 <td>
 Analog Output Module
 </td>
 <td>
 4-20mA; 500 Ohms max load
 </td>
 </tr>
 <tr>
 <td>
 Digital Input/Output Module
 </td>
 <td>
 Configurable as Frequency or Digital I/O
 </td>
 </tr>
 <tr>
 <td>
 Frequency Output
 </td>
 <td>
 0 to 10KHz proportional to flow rate
 </td>
 </tr>
 </tbody>
 </table>
 </h4><h4>&nbsp;</h4><h4>
 </h4><h4>&nbsp;</h4><h4>
 </h4><h4>Physical Specifications</h4><h4>
 <table class="table table-bordered">
 <tbody>
 <tr>
 <td>
 Process Connections
 </td>
 <td>
 3/8” tube connection **
 </td>
 </tr>
 <tr>
 <td>
 Wetted Material
 </td>
 <td>
 PFA
 </td>
 </tr>
 <tr>
 <td>
 Sensor Dimensions
 </td>
 <td>
 L(254mm) x W(73mm) x H(115mm)
 </td>
 </tr>
 <tr>
 <td>
 Transmitter Dimensions
 </td>
 <td>
 L(210mm) x W(58.5mm) x H(133.5mm)
 </td>
 </tr>
 <tr>
 <td>
 Weight
 </td>
 <td>
 Sensor: 0.86 Kg; Transmitter: 1.22 Kg
 </td>
 </tr>
 <tr>
 <td>
 Cable Length
 </td>
 <td>
 Up to 30m max cable length between sensor and electronics assemblies
 </td>
 </tr>
 </tbody>
 </table>
 ** Consult factory for other types of process connection requirements.
 &nbsp;
 <em>For additional specification details, please refer to the product&nbsp;data sheet located in the documentation tab of this web page.&nbsp;</em> </h4>    