<h4>Features</h4>
 <ul>
 <li> <i class="fa fa-check"></i> Pressure measurement of any gas, liquid or oil (except substances which may corrode stainless steel 316L)</li>
 <li> <i class="fa fa-check"></i> Auto shift function: with change in the original pressure, the external input adjusts the determined level to match the change in pressure (only available in models with auto shift/hold function)</li>
 <li> <i class="fa fa-check"></i> High display resolution:</li>
 <ul>
 <li> <i class="fa fa-check"></i>
  - negative pressure: 0.1 kPa</li> 
  <li> <i class="fa fa-check"></i> - standard pressure: 0.1 kPa, 1 kPa </li>
  <li> <i class="fa fa-check"></i>- compound pressure: 0.1 kPa</li></ul>
 <li> <i class="fa fa-check"></i> Two independent outputs (N.O. / N.C. output selectable)</li>
 <li> <i class="fa fa-check"></i> Hold function: hold current display value or control output</li>
 <li> <i class="fa fa-check"></i> High/low-limit display scale function</li>
 <li> <i class="fa fa-check"></i> Forced output control mode for device testing and maintenance</li>
 <li> <i class="fa fa-check"></i> One-touch connector type for easy wiring and maintenance</li>
 <li> <i class="fa fa-check"></i> Analog output: voltage (1-5 VDC), current (DC 4-20 mA)</li>
 <li> <i class="fa fa-check"></i> Zero-point adjustment function, peak value monitoring function, chattering prevention function</li>
 </ul>