<table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Differential Pressure - Piezoresistive</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  0 to 300 hPa
  </td>
  </tr>
  <tr>
  <td>Accuracy</td>
  <td>
  ±0.5 hPa (0.0 to +50.0 hPa)
  ±1 % of mv (+50.1 to +100.0 hPa)
  ±1.5 % of mv (Remaining Range)
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  0.1 hPa
  </td>
  </tr>
  </tbody>
  </table>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Flue gas O?</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  0 to 21 Vol.%
  </td>
  </tr>
  <tr>
  <td>Accuracy</td>
  <td>
  ±0.2 Vol.%
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  0.1 Vol.%
  </td>
  </tr>
  </tbody>
  </table>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Flue gas CO (with H?-compensation)</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  0 to 8000 ppm
  </td>
  </tr>
  <tr>
  <td>Accuracy</td>
  <td>
  ±10 ppm or ±10 % of mv (0 to 200 ppm)
  ±20 ppm or ±5 % of mv (201 to 2000 ppm)
  ±10 % of mv (2001 to 8000 ppm)
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  1 ppm
  </td>
  </tr>
  </tbody>
  </table>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">CO determination (with H?-compensation), automatic dilution</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  0 to 30000 ppm
  </td>
  </tr>
  <tr>
  <td>Accuracy</td>
  <td>
  ±200 ppm or ±20 % of mv (0 to 30000 ppm)
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  1 ppm
  </td>
  </tr>
  </tbody>
  </table>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Flue gas NO</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  0 to 3000 ppm
  </td>
  </tr>
  <tr>
  <td>Accuracy</td>
  <td>
  ±5 ppm (0 to 100 ppm)
  ±5 % of mv (101 to 2000 ppm)
  ±10 % of mv (2001 to 3000 ppm)
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  1 ppm
  </td>
  </tr>
  </tbody>
  </table>
  <p>Option</p>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Flue gas Draught</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  -9.99 to +40 hPa
  </td>
  </tr>
  <tr>
  <td>Accuracy</td>
  <td>
  ±0.02 hPa or ±5 % of mv (-0.50 to +0.60 hPa)
  ±0.03 hPa (+0.61 to +3.00 hPa)
  ±1.5 % of mv (3.01 to +40.00 hPa) (the greater value applies)
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  0.01 hPa
  </td>
  </tr>
  </tbody>
  </table>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Temperature</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  -40 to +1200 °C
  </td>
  </tr>
  <tr>
  <td>Accuracy</td>
  <td>
  ±0.5 °C (0.0 to +100.0 °C)
  ±0.5 % of mv (Remaining Range)
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  0.1 °C (-40 to +999.9 °C)
  1 °C (Remaining Range)
  </td>
  </tr>
  </tbody>
  </table>
  <p>dependent on the thermocouple</p>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Flue gas degree of effectivity, Eta (calculated)</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  0 to 120 %
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  0.1 %
  </td>
  </tr>
  </tbody>
  </table>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Flue gas loss (calculated)</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  0 to 99.9 %
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  0.1 %
  </td>
  </tr>
  </tbody>
  </table>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">Flue gas CO? calculation (calculated from O?)</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Measuring range</td>
  <td>
  Display range 0 to CO? max
  </td>
  </tr>
  <tr>
  <td>Accuracy</td>
  <td>
  ±0.2 Vol.%
  </td>
  </tr>
  <tr>
  <td>Resolution</td>
  <td>
  0.1 Vol.%
  </td>
  </tr>
  </tbody>
  </table>
  <table class="table table-bordered">
  
  <thead>
  <tr>
  <th colspan="2">General technical data</th>
  </tr>
  </thead>
  <tbody>
  <tr>
  <td>Dimensions</td>
  <td>
  270 x 160 x 57 mm
  </td>
  </tr>
  <tr>
  <td>Operating temperature</td>
  <td>
  -5 to +45 °C
  </td>
  </tr>
  <tr>
  <td>Warranty</td>
  <td>
  Instrument: 48 months; Probe: 48 months; Gas sensors (O?, CO) 48 months; NO sensor 24 months; Thermocouple 12 months; rechargeable battery: 12 months
  </td>
  </tr>
  <tr>
  <td>Power supply</td>
  <td>
  Rech. batt. block 3.7 V / 2.6 Ah; Mains unit 6 V / 1.2 A (optional)
  </td>
  </tr>
  <tr>
  <td>System requirements</td>
  <td>
  requires iOS 7.1 or newer; requires Android 4.3 or newer; requires mobile end device with Bluetooth 4.0
  </td>
  </tr>
  <tr>
  <td>Maximum</td>
  <td>
  500000 readings
  </td>
  </tr>
  <tr>
  <td>Storage temperature</td>
  <td>
  -20 to +50 °C
  </td>
  </tr>
  <tr>
  <td>Weight</td>
  <td>
  720 g (excluding battery)
  </td>
  </tr>
  </tbody>
  </table>