<h4>Performance Specifications</h4>
 <table class="table table-bordered">
 <tbody>
 <tr>
 <td >
 <p>Standard Full Scale Range:</p>
 </td>
 <td >
 <p>50, 100, 250, 500, 1000, 1500, 2500*, 4000, 8000, 12000 mL/min</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Accuracy**: <br>(for room temperature DIW)</p>
 </td>
 <td >
 <p>±1% of set point or ± 3mL/min<br>(whichever is larger)</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Repeatibility**:</p>
 </td>
 <td >
 <p>±1% of set point or ±1 mL/min<br> (whichever is larger)</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Control Repeatibility:</p>
 </td>
 <td >
 <p>±0.5% of set point or ±0.5mL/min<br>(whichever is larger)</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Flow Control Time:</p>
 </td>
 <td >
 <p>&lt; 3 sec</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Fluid Temperature:</p>
 </td>
 <td >
 <p>10 - 60 ºC***</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Ambient: Temperature/Humidity:</p>
 </td>
 <td >
 <p>0 - 40 ºC / 30 - 80% RH, without Dew</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Maximum Expected Operating Pressure:</p>
 </td>
 <td >
 <p>50 psig</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Maximum Safe Internal Pressure:</p>
 </td>
 <td >
 <p>70 psig</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Differential Pressure Range:</p>
 </td>
 <td >
 <p>7 to 30 psid</p>
 </td>
 </tr>
 </tbody>
 </table>
 <p>The enclosure footprint may be larger for these flow ranges to meet the pressure drop specification.<br>The minimum differential pressure requirements can be higher for these ranges. <br>**Please consult with Malema for tighter accuracy/repeatability needs. <br>*** Consult the factory for higher temperature application</p>
 <p>&nbsp;</p>
 <h3>Material Specifications</h3>
 <table class="table table-bordered">
 <tbody>
 <tr>
 <td >
 <p>Non Wetted Parts, Enclosure:</p>
 </td>
 <td >
 <p>PPS, PEEK, Acrylic,Vinyl, PVC**</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Wetted parts:</p>
 </td>
 <td >
 <p>PFA, PTFE, Pt cured Silicone*</p>
 </td>
 </tr>
 </tbody>
 </table>
 <p>*Only used in the Slurry Module<br>**Flame retardant (FMET4325)</p>
 <table class="table table-bordered">
 <tbody>
 <tr>
 <td >
 <p>Mounting Orientation:</p>
 </td>
 <td >
 <p>Horizontal or Vertical</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Fluid Connections:</p>
 </td>
 <td >
 <p>Inlet/Outlet:1/4” or 3/8”, Flare or Pillar</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Flow Restrictions (orifice):</p>
 </td>
 <td >
 <p>&gt; 2 mm</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Ingress Rating:</p>
 </td>
 <td >
 <p>IP64</p>
 </td>
 </tr>
 </tbody>
 </table>
 <p>&nbsp;</p>
 <h3>Electrical Specifications</h3>
 <table class="table table-bordered">
 <tbody>
 <tr>
 <td >
 <p>Power Supply Input:</p>
 </td>
 <td >
 <p>24 Vdc ± 10%</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Current Consumption:</p>
 </td>
 <td >
 <p>Max 0.5 A</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Alarm Signals:</p>
 </td>
 <td >
 <p>Max 30 Vdc, 200 mA NPN open collector</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>ControlSignal In*:</p>
 </td>
 <td >
 <p>0 to 10 Vdc or 4 to 20 mA</p>
 </td>
 </tr>
 <tr>
 <td >
 <p>Flow Signal Out*:</p>
 </td>
 <td >
 <p>0 to 10 Vdc or 4 to 20 mA**</p>
 </td>
 </tr>
 </tbody>
 </table>
 <p>* Other options available<br>** Both Active and Passive current options available</p>
 <p>&nbsp;</p>
 <p><em>For additional specification details, please refer to the product&nbsp;data sheet located in the documentation tab of this web page.&nbsp;</em></p>